Price on the app only:
Special discount has ended
Please note each user for only 1 item(s)
if the goods arrival in 90 days,we will inform you by email!
Error appeared,Please try again later!
|Sample Rate||Real-Time Sample: 1GS/s|
|Normal||Normal data only|
|Peak Detect||High-frequency and randon glith capture|
|Average||Wavefom Average, selectable 4,8,16,32,64,128|
|Inputs Coupling||AC, DC, GND|
|Inputs Impendance||1MΩ±2% ‖20pF±3pF|
|Probe Attenuation||1X, 10X|
|Supported Probe Attenuation Factor||1X, 10X, 100X, 1000X|
|Maximum Input Voltage||CAT I and CAT II: 300VRMS (10×), Installation Category;
CAT III: 150VRMS (1×);
Installation Category II: derate at 20dB/decade above 100kHz to 13V peak AC at 3MHz* and above. For non-sinusoidal waveforms, peak value must be less than 450V. Excursion above 300V should be of less than 100ms duration. RMS signal level including all DC components removed through AC coupling must be limited to 300V. If these values are exceeded, damage to the oscilloscope may occur.
|Sample Rate Range||500MS/s--1GS/s|
|Waveform Interpolation||(sin x)/x|
|SEC/DIV Range||4ns/div to 40s/div|
|Sample Rate and
Delay Time Accuracy
|±50ppm(at over any ≥1ms time interval)|
|Position Range||20ns/div to 80us/div; (-8div x s/div) to 40ms;
200us/div to 40s/div; (-8div x s/div) to 400s
|Delta Time Measurement Accuracy
|Single-shot, Normal mode:± (1 sample interval +100ppm × reading + 0.6ns);
>16 averages:± (1 sample interval + 100ppm × reading + 0.4ns);
Sample interval = s/div ÷ 200
|Vertical Resolution||8-bit resolution, all channel sampled simultaneously|
|Position Range||2mV/div to 10V/div|
|Rise Time at BNC( typical)||3.5ns|
|Analog Bandwidth in Normal and Average modes at BNC or with probe, DC Coupled||2mV/div to 20mV/div, ±400mV; 50mV/div to 200mV/div, ±2V
500mV/div to 2V/div, ±40V; 5V/div, ±50V
|Math||+, -, *, /, FFT|
|FFT||Windows: Hanning, Flatop, Rectamgular, Bartlett, Blackman;
1024 sample point
|Low Frequency Response (-3db)||≤10Hz at BNC|
|DC Gain Accuracy||±3% for Normal or Average acquisition mode, 5V/div to 10mV/div;
±4% for Normal or Average acquisition mode, 5mV/div to 2mV/div
|DC Measurement Accuracy,
Average Acquisition Mode
|When vertical displacement is zero, and N ≥16:± (3% × reading + 0.1div + 1mV) only 10mV/div or greater is selected;
When vertical displacement is not zero, and N≥16: ± [3% × (reading + vertical position) + 1% of vertical position + 0.2div]; Add 2mV for settings from 2mV/div to 200mV/div; add 50mV for settings from 200mV/div to 5V/div
|Volts Measurement Repeatability,
Average Acquisition Mode
|Delta volts between any two averages of ≥16 waveforms acquired under same setup and ambient conditions|
|Trigger Types||Edge, Video, Pulse, Slope, Over time, Alternative|
|Trigger Source||CH1, CH2, EXT, EXT/5, AC Line|
|Trigger Modes||Auto, Normal, Single|
|Coupling Type||DC, AC, Noise Reject, HF Reject, LF Reject|
(Edge Trigger Type)
1div from DC to 10MHz; 1.5div from 10MHz to 100MHz; 2div from 100MHz to Full;
200mV from DC to 100MHz; 350mV from 100MHz to 200MHz;
1V from DC to 100MHz;1.75V from 100MHz to 200MHz;
AC: Attenuates signals below 10Hz;
HF Reject: Attenuates signals above 80kHz
LF Reject: Same as the DC-coupled limits for frequencies above 150kHz; attenuates signals below 150kHz
|Trigger Level Range||CH1/CH2: ±8 divisions from center of screen;
|Trigger Level Accuracy( typical)Accuracy is for signals having rise and fall times ≥20ns||CH1/CH2: 0.2div × volts/div within ±4 divisions from center of screen;
EXT: ± (6% of setting + 40mV);
EXT/5: ± (6% of setting + 200mV);
|Set Level to 50%(typical)||Operates with input signals ≥50Hz|
|Video Trigger Type||CH1, CH2: Peak-to-peak amplitude of 2 divisions;
|Signal Formats and Field Rates, Video Trigger Type||Supports NTSC, PAL and SECAM broadcast systems for any field or any line|
|Holdoff Range||100ns ~ 10s|
|Pulse Width Trigger|
|Pulse Width Trigger Mode||Trigger when (< , >, = , or ≠); Positive pulse or Negative pulse|
|Pulse Width Trigger Point||Equal: The oscilloscope triggers when the trailing edge of the pulse crosses the trigger level.
Not Equal: If the pulse is narrower than the specified width, the trigger point is the trailing edge. Otherwise, the oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width.
Less than: The trigger point is the trailing edge.
Greater than (also called overtime trigger): The oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width
|Pulse Width Range||20ns ~ 10s|
|Slope Trigger Mode||Trigger when (< , > , = , or ≠ ); Positive slope or Negative slope|
|Slope Trigger Point||Equal: The oscilloscope triggers when the waveform slope is equal to the set slope.
Not Equal: The oscilloscope triggers when the waveform slope is not equal to the set slope.
Less than: The oscilloscope triggers when the waveform slope is less than the set slope.
Greater than: The oscilloscope triggers when the waveform slope is greater than the set slope.
|Time Range||20ns ~ 10s|
|Over Time Mode||Rising edge or Falling edge|
|Time Range||20ns ~ 10s|
|Trigger on CH1||Internal Trigger: Edge, Pulse Width, Video, Slope|
|Trigger on CH2||Internal Trigger: Edge, Pulse Width, Video, Slope|
|Trigger Frequency Counter|
|Readout Resolution||6 digits|
|Accuracy (typical)||±30ppm (including all frequency reference errors and ±1 count errors)|
|Frequency Range||AC coupled, from 4Hz minimum to rated bandwidth|
|Signal Source||Pulse Width or Edge Trigger modes: all available trigger sources
The Frequency Counter measures trigger source at all times, including when the oscilloscope acquisition pauses due to changes in the run status, or acquisition of a single shot event has completed.
Pulse Width Trigger mode: The oscilloscope counts pulses of significant magnitude inside the 1s measurement window that qualify as triggerable events, such as narrow pulses in a PWM pulse train if set to < mode and the width is set to a relatively small time.
Edge Trigger mode: The oscilloscope counts all edges of sufficient magnitude and correct polarity.
Video Trigger mode: The Frequency Counter does not work.
|Cursor Measurement||Voltage difference between cursors: △V
Time difference between cursors: △T
Reciprocal of △T in Hertz (1/ΔT)
|Auto Measuerment||Frequency, Period, Mean, Pk-Pk, Cycli RMS, Minimum, Maximum, Rise time, Fall Time, +Pulse Width, -Pulse Width, Delay1-2Rise, Delay1-2Fall, +Duty, -Duty, Vbase, Vtop, Vmid, Vamp, Overshoot, Preshoot, Preiod Mean, Preiod RMS, FOVShoot, RPREShoot, BWIDTH, FRF, FFR, LRR, LRF, LFR, LFF|
|Display Type||7 inch 64K color TFT (diagonal liquid crystal)|
|Display Resolution||800 horizontal by 480 vertical pixels|
|Display Contrast||Adjustable (16 gears) with the progress bar|
|Probe Compensator Output|
|Output Voltage( typical)||About 5Vpp into ≥1MΩ load|
|Supply Voltage||100-120VACRMS(±10%), 45Hz to 440Hz, CATⅡ
120-240VACRMS(±10%), 45Hz to 66Hz, CATⅡ
|Fuse||2A, T rating, 250V|
|Temperature||Operating: 32℉ to 122℉ (0℃ to 50℃);
Nonoperating: -40℉ to 159.8℉ (-40℃ to +71℃)
|Humidity||+104℉ or below (+40℃ or below): ≤90% relative humidity;
106℉ to 122℉ (+41℃ to 50℃): ≤60% relative humidity
|Altitude||Operating: Below 3,000m (10,000 feet);
Nonoperaring: Below 15,000m(50,000 feet)
|Size||Length 315mm, Width 130mm, Height 152mm|
|Weight||2.9KG(with Packing); 2.00KG(without Packing）|
Questions & Answers
Dear buyer, Good day. Thank you for your ask. This scope DSO5102P is works with win 10. Thanks and a nice day. … see more Dear buyer, Good day. Thank you for your ask. This scope DSO5102P is works with win 10. Thanks and a nice day. withUtrade see less
By withUtrade SELLER on March 26, 2018
Dear buyer, Good day. Thank you for your ask. This scope has not battery inside, only for AC power. Hopefully this answer is helpful. Thanks and have a nice day.… see more Dear buyer, Good day. Thank you for your ask. This scope has not battery inside, only for AC power. Hopefully this answer is helpful. Thanks and have a nice day. withUtrade see less
By withUtrade SELLER on March 10, 2018
Dear buyer, Good day. Thank you for your ask. This machine is 110-240V. So no matter 220V or 110V, it all works. If you still have any questions, please don't hesitate to ask us. Hopefully this answer is helpful.… see more Dear buyer, Good day. Thank you for your ask. This machine is 110-240V. So no matter 220V or 110V, it all works. If you still have any questions, please don't hesitate to ask us. Hopefully this answer is helpful. Thanks and have a nice day. withUtrade see less
By withUtrade SELLER on January 11, 2018
Not sure... I mainly use it for digital. It is a nice scope and has been a valuable tool.
By TerrysOpinion on January 2, 2018
It can't test above 5khz
By JR TECH SELLER on October 12, 2017
By preston m. woodard on September 17, 2017
Dear Buyer, Good day. Thank you for your ask. About your question, the front USB port is for save date and upgrade the system. And the back USB port is for connect the PC software. Hopefully this answer is helpful and if you still have any questions, please let us know. :) Thanks and have a nice day.… see more Dear Buyer, Good day. Thank you for your ask. About your question, the front USB port is for save date and upgrade the system. And the back USB port is for connect the PC software. Hopefully this answer is helpful and if you still have any questions, please let us know. :) Thanks and have a nice day. withUtrade Team see less
By withUtrade SELLER on July 19, 2017
Yes. There is a switch on the probe to go between 1 x and 10 x.
By chuckb on July 10, 2017
Dont know about ce standard. It just works well.
By Dai Vu on June 12, 2017
Please note that TOMTOP Forums are a community for all communicating and getting help each other. There will be some enthusiastic friends participate in your discussions. Of course TOMTOP customer service (with tomtop customer service icon) response is guaranteed ,which is the same way you contact us at http://www.tomtop.com/contactsopen this forum in full page
based on 12 Customer Reviews
Share your thoughts with other customers and get Tomtop Points, the first 5 reviews get DOUBLE Tomtop Points!Write a Customer Review
December 28, 2017
Produit conforme et de qualité très correct. Cherchant depuis quelques temps un oscillo pour pas trop cher, je me suis penché sur ce modèle. Et pour le moment je ne le regrette pas. L'écran est grand et lisible, l’ergonomie des menus et boutons me satisfont. Facilement portable, autonome (pas comme un modèle USB), rapide et pas trop cher (Actuellement 236.99€). Cet appareil a tout pour lui. Je regrette le ventilo qui fait un léger bruit. Pourquoi ne pas avoir monter un bon radiateur, ça aurait été mieux. Pilotable par USB et possibilité de copier l'écran sur une clef USB sont un vrai plus. Si il résiste au temps (surement pas autant que les modèles a tube, malheureusement) c'est une très bonne affaire.
By Alain BASTY
December 15, 2017
Cons - Le titre est trompeur, cependant la description et la photo sont corrects (on voit clairement le modèle sur la photo) - La touche F4 sur mon modèle est défaillante, je m'en sors en exerçant un appui sur le côté droit de la touche avec l'ongle - On le trouve moins cher ailleurs... Pros - Simple d'utilisation, self tests et auto calibration OK - Documentation OK à télécharger sur le site de Hantek comme les drivers et apps PC (pas tester) - Convient parfaitement à mon utilisation (petite électronique, circuit logique, etc.)
June 1, 2018
April 5, 2018
February 2, 2018
By Daffy Paris
December 29, 2017
By Loris Zoppelletto
December 21, 2017
Ero timoroso di trovarmi con una cineseria poco affidabile, visto il basso prezzo, invece ho avuto tutto quello che mi aspettavo per il mio laboratorio di elettronica. Ha un generatore di segnale dignitoso con forma d'onda importabile e l'oscilloscopio ha una buona qualità di risoluzione del segnale per utilizzi amatoriali. Anche le sonde sono di buona qualità. Un ottimo acquisto nel complesso
December 13, 2017
By Mezzena C.
December 8, 2017
Ottimo veramente per la sua fascia di prezzo. Molto soddisfatto dell'acquisto. Menu molto completi e molto intuitivi. Usato subito senza necessità di leggere il manuale. Consegnato in tempi abbastanza rapidi considerato che è stato spedito dalla Cina. Lo uso da circa 1 settimana abbastanza pesantemente ed ha sempre risposto perfettamente. Lo consiglio a chiunque non abbia bisogno di misurazioni molto particolari.
By kevin brisset
November 26, 2017
November 7, 2017
January 9, 2016
Hot Event More
Top Sellers More
Your question has been submitted successfully
Please activate your account first. We will resend the activation email to help you complete your registration.
OUR SITE RUNS OF COOKIES