Price on the app only:
Special discount has ended
Please note each user for only 1 item(s)
if the goods arrival in 90 days,we will inform you by email!
Error appeared,Please try again later!
|Key Features||Frequency Range||Sine Wave||
Normal Modes(Optional): 0.01Hz~6MHz / 0.01Hz~12MHz / 0.01Hz~20MHz / 0.01Hz~25MHz
|TTL Digital Signal Wave||0.01Hz~6MHz|
|Output Modulation||Frequency Sweep|
|Waveform Types||Sine/Triangle/Square/Lifting Sawtooth Wave, CMOS Digital Signal Wave and Arbitrary Waveform|
|Waveform Amplitude Resolution||12 Bits|
|The Min. Frequency Resolution||10mHz|
|Amplitude Range(Peak to Peak)||
1mVp-p (-20dB Attenuation)
10mVp-p (No Attenuation)
|Offset Range||-120%~120% (The ratio of bias voltage and signal amplitude)|
|Sine Wave||Harmonic Arrived System||40dBc(<1MHz); 35dBc(1MHz~20MHz)|
|Square Wave||Lifting Along Time||≤20ns|
|Duty Cycle Adjustment Range||0%~99.9%|
|TTL||Lifting Along Time||≤20ns|
|Scan||Scan Mode||Linear Sweep, Log Sweep|
|Scan Range||It is determined by the sweep parameter settings|
|External Measurements||Frequency Measuring Range||
GATE-TIME=10S 0.1Hz - 60MHz
GATE-TIME=1S 1Hz - 60MHz
GATE-TIME=0.1S 10Hz - 60MHz
GATE-TIME=0.01S 100Hz - 60MHz
Input Voltage Range
|Positive and Negative Pulse Width Measurement||10ns Resolution, the Max. Measurable 10s|
|Periodic Measurement||20ns Resolution, the Max. Measurable 20s|
|Duty Cycle Measurement||0.1% Resolution, Measuring Range from 0.1%~99.9%|
|Source Selection||1. Ext. IN Input(AC Signal) ; 2. TTL-IN Input(Digital Signal)|
|Interface||Interface Mode||using USB to Serial Interface|
|Protocol||Using the Command Line, the Agreement Public|
|Power Supply||DC||MHS5200A DC 5V Power Supply|
|Size||19.5 * 17.5 * 7cm / 7.67 * 6.9 * 2.75in|
|Weight||542g / 19.12oz|
|Package Size||24 * 21 * 9.5cm / 9.45 * 8.3 * 3.74in (L * W * H)|
|Package Weight||986g / 34.78oz|
Questions & Answers
Please note that TOMTOP Forums are a community for all communicating and getting help each other. There will be some enthusiastic friends participate in your discussions. Of course TOMTOP customer service (with tomtop customer service icon) response is guaranteed ,which is the same way you contact us at http://www.tomtop.com/contactsopen this forum in full page
based on 52 Customer Reviews
Share your thoughts with other customers and get Tomtop Points, the first 5 reviews get DOUBLE Tomtop Points!Write a Customer Review
By AUgie the Prospector
January 23, 2018
All I've done is try out the built-in capabilities, just to be sure it wasn't DOA, and was a version with the advertised features. Everything works as expected. Any shortcomings in signal quality are minor and expected given the price point. I connected it to the Siglent SDS1102CML digital scope for these tests, and was reminded how easy that oscilloscope is to use as well. First, anyone interested in learning about this signal generator will find a lot more online about the model MHS-5200A, which appears to be the exact same unit, under a different name. There are reviews and upgrades, including fixing the over-bright display (a minor issue) and correcting some of the distortion in the sine wave output (how important this is depends on how spectrally pure an output you need). There is also some aftermarket software hobbyists have developed for uploading arbitrary waveforms, if the supplied software doesn't meet your needs. Some people have even managed to "close the loop" and download waveforms captured on their digital storage scopes (like the one I use) into their computer, edit them, and then upload them into the signal generator. This is way more than I need - I'll likely never attach it to a computer - but some users of this device will doubtless want to follow these brave hardware hackers' lead. A nice English manual is available from one of the serious hobbyist sites, and maybe other places too. It was written by an English-speaking hobbyist based on the Chinese manual and online sources - it covers all the basic internal functions with clarity. I don't like posting links, so just Google it yourself. Search for MHS5200.pdf. Follow a result on eevblog.com about reverse engineering the serial protocol, look at post number 18 and you'll see a link to the pdf file on github. It scans clean of malware and is very clearly written. However you find it, it will be useful. While I didn't have my scope analyze for spectral purity (see MHS-5200A YouTube videos for that), I did run the generator through its built in paces. As others have noted, the square wave begins losing its crispness much above 1 MHz, and is pretty "round" by the advertised 6 MHz. Sine waves look pretty good all the way to 25 MHz. Locking the two channels works, all the wave shapes work. Adjusting the duty factor of square waves works. Adjusting the relative phases of the two channels works. Sweeping the frequency works as advertised. The input frequency counter works in its various frequency, period, duty factor, etc., modes, including when feeding back one of the output channels into it, set up prior to moving to the signal input part of the menu. (Not that anyone with a modern digital oscilloscope needs this feature.) I didn't test the rear panel TTL inputs or outputs. I also didn't try saving a setup of the controls into one of the internal memories (probably works). Adjusting the amplitude works, but I forgot to try the offset - I'm sure it works too. I noticed that the square wave has a small "back porch" (slight drop in level near the trailing edge), even at low frequencies, but is within the range of expected signal impurities and would just about never be an issue. Amplitudes and frequencies agreed extremely well between the signal generator and my digital scope. Conclusion; This seems to be the exact same device as the more commonly discussed MHS-5200A, and all the features and functions that I tested work as expected (none involving a computer were tried). It is a great value for what it can do and has even more potential when interfaced to a computer, or if modified for greater spectral purity. Amazon seems to be selling, in the KKmoon, an MHS-5200A under a different name, for a very good price. I saw no issues with the build quality or construction of my unit. Finally, I didn't find the controls as hard to fathom as I expected. They can be a little slow to navigate, but with a few minutes use they seemed fairly natural and required no reference to the manual. The English manual did come in handy preventing a lot of guessing at a few points, so I do recommend it highly.
By Radio Guy
December 23, 2017
This is an inexpensive signal generator that offers a lot for the low price. While expensive test gear will outperform this - for many uses this generator is really handy. Before I purchased this I checked out videos on you tube and was generally impressed with what you could do with an inexpensive piece of test gear. Some of the videos showed some ring on a square wave from the generator. The model I received did not have this problem so I suspect that the generator design may have been touched up to take care of this small issue. Overall the generator give the hobbyist a lot of capability. For my use it's been great.
By S. Thompson
September 15, 2017
First thing I noticed It was easy to set, Its the easiest to set of My other frequency generators but that Is once You know how to set It, there was some things right out of the box but there were some things I would have Not been able to figure out, I had to go watch some videos to figure It out. To adjust the level at the top of the screen click the bottom left button then push the top right button and do It again to go to the bottom of the screen. To switch channels click the bottom left button then the top left button, To start click the bottom left button then the bottom right button, You must ad just the settings as the frequency before It will start, I thought I got a defective one because I did Not know that. The amplitude power output will Not go to zero volts It goes to from 00.20 volts to 13 volts but It will read 20 volts. Also for the program to work You must have a newer Microsoft Operating System It would Not work with My old Windows XP A quick history about Doctor: Rife cured 16 out of 16 People of cancer In 1934 using am radio frequencies waves through a plasma bulb, but a business partner tampered with the frequencies and They were selling machines that did Not work which gave big drug companies a chance to put Them out of business and In the 1950's Doctor: Crane wanted to reinvent the Rife machine but using Audio frequencies and Tens pads which did Not work as well as the original Rife machine. I bought this because I wanted a 3.3 MHz carrier wave for a Rife machine but I did Not know how easy It Is to adjust. This will work as a Doctor: Clark zapper or Croft zapper It has adjustable voltage up to 13 volts output though It will read 20 volts output the screen shows AMPL short for amplitude voltage that Is the voltage output adjustment. I do Not think It has enough voltage output for Tens pads or any Non metal connection, but It has enough voltage out put If used with metal as Copper or Stainless Steel, Using Copper handles I could feel the voltage at 6 volts and 12 volts I was getting a good zapping and what I always say If You can Not feel It shocking You the Its probably Not doing anything to the pathogens. Also Doctor: Clark said to use a positive offset, on screen says OFFS This can also be used as the frequency generator In the Rife/Doug coil machine, A Technician In a video said this has a ring In the Square wave and put In better parts but I do Not know how much that will effect It. The Spooky2 software program will Not work with this Generator, It did Not detect It, but years a go We were able to out smart that Internet virus company (You know that big company that puts their big genitals where It does Not belong and freezes up Your account for No understandable reason) and I bought a Spook2 generator but became frustrated trying to adjust the settings I tapped the screen with My finger It and broke the LCD screen but the Spooky2 program still recognised It and asked to look for more generators but I have Not tried It yet. Also the Spooky2 booster does Not fit this Generator but all It does Is link the 2 channels together and has a 103 resistor for making Colloidal Silver. It will Not work as a Doctor: Beck, Blood Electrification unit or a Tens machine, both Tens machines and the Blood Electrification unit Is around 50 volts AC output but the Blood Electrification unit Is set around 4 Hz and Is higher amperage than Tens machines, that Is why Tens machines are everywhere at low prices because they do Not have enough amperage to do anything a Tens machine will shock You but does Not even have enough amperage to light a LED bulb...........
September 14, 2017
Seems to be quite accurate, checked with my Digital Oscope. It comes with a piece of software for Windows. With the software, connected through USB to a Windows computer, it is a lot easier to set parameters. On the other hand, the computer has to be running for the software to be used, duh. Of course! There is however a flaw / bug in the device firmware during power on. At times when powered on, the display is corrupted and so are the control functions. Power off and on again has fixed it in the past. Otherwise, at the price to performance ratio, it is quite usable for some bench projects. I don't know why it sometimes doesn't initializes at first power on. From my own experience in programming micro controller, I guess there isn't enough delay to initialize the micro.
By Mark T.
September 12, 2017
Hardware Great for the Price, but need to have better software sent with the Unit on the small disc. Missing main program and you have to read through all the reviews on this sight to get to the point you can program the unit via the USB cable. A bit shy on documentation but again reading through these reviews and you will be able to get by. It may not directly answer all your questions, but you will come away with enough to be able to use most all of its features. Again it is a very good unit for the price. So because of the poor / missing files I deducted 20% and could only offer 4 stars.
By L. F. Lederer
September 9, 2017
It's darn good. The signals are crisp and clear on my analog scope. The amplitude settings seem dead on. The software is a big plus (must), making it very easy to setup and designing arbitrary waveforms which adds some flexibility to the unit. There is third party software available as well for developing your even more arbitrary waveforms. I also liked that you can output to both channels at the same time and with totally different waveforms and frequencies.It also came with a handy TTL interface that plugs into the back of the unit. As stated in an other review the 25 MHz model only puts out a sine wave at that frequency. All other waveforms are limited to around 6 MHz before they start to distort. And at 25 MHz they all looked like a sine wave. The frequency counter seems to work OK but I don't have a calibrated device to measure its accuracy. Three negatives. One the manual is definitely lacking clarity on the operation and functionality of the instrument. Two, the software on the CD was corrupted and it took me nearly a day to find their website, no thanks to the seller in helping me. It took three tries to download the 110 MB file and nearly two hours on the final try as their download speed averaged about 350 kbps (sic). Three, selecting a waveform, frequency, retrieving or storing to memory and selection of other functions is clunky without the software. Some of the buttons have dual functions so you have to push a "shift" button first to access the second function. They also located the on/off switch on the back of the unit. Overall, for under a $100 it will meet my needs, I don't work in a lab or calibrate instruments but as an engineer it does a lot for the money.
January 30, 2018
By Martin Peck
January 25, 2018
January 20, 2018
By Kimberly N. Marshall
January 17, 2018
January 7, 2018
By Mike M
January 5, 2018
December 31, 2017
December 26, 2017
What an amazing use of micro-controllers for direct signal generation. A bit cheesy use of just 6 click buttons, but a real nice encoder knob for the many adjustable parameters and features up the wahzoo.. Locating the (large) knob at the far right is the best ergonomic location if you ask me. I've been working in electronics since age 13, that's lots of years ago and I know great engineering when I see it.
By Jesse L.
December 19, 2017
By Emily M Lopez
December 13, 2017
September 15, 2017
July 19, 2017
December 21, 2015
December 20, 2015
Hot Event More
Top Sellers More
Your question has been submitted successfully
Please activate your account first. We will resend the activation email to help you complete your registration.